The operation manuals on this webpage are the most current versions available at this time. All manuals are subject to change without notice. To ensure that you always have the most current manual, please check this website regularly for updates.
Remove the air scaler from the coupling joint and detach the tip.
Bring air scalers to a well-prepared cleaning room without contamination.
Remove organic contaminants with a paper wipe.
Manual: Clean the external surface of air scaler with running water
Automated: NSK air scalers with this symbol are the thermo-disinfector-ready and may be washed and disinfected using a thermo-desinfector.
When using a thermo-desinfector, be sure to follow its manufacturer's instructions.
Manual: Wipe the external surface of air scalers with cleaning or alcohol for disinfection.
Essential after every thermodisinfection process
Essential prior to each sterilisation process
Manual: Supply 1-2 drops of NSK Turbine Oil to the drive air port of the air scalers. Connect to the coupling joint and operate for 5 seconds. Note: Excessive oil may cause instability in instrument rotation. Please consult the operation manual before use.
Place air scalers in a sterilisation pouch and seal them.
EN13060 4.6.3 recommends autoclaving for 3 minutes (minimum holding time) at 134℃ or 15 minutes (minimum holding time) at 121℃ .
NSK recommends Class B or S sterilisation.
Immediately after a sterilisation cycle, remove air scalers from the autoclave.
Keep them in a Dust-free sterilised storage or bring them to a treatment room for use.